The DECA connects to the SAT input, so what needs to be measured is the SAT input [center pin to shield] with the DECA removed.
Would be beneficial for our members to describe the process ? Would it require to open DECA box and measure the voltage inside of it?
Welcome to DBSTalk
Like most online communities you must register to view or post in our community. Sign-up is a free and simple process that requires minimal information. Be a part of our community by signing in or creating an account. The Digital Bit Stream starts here!
- Reply to existing topics or start a discussion of your own
- Subscribe to topics and forums and get email updates
- Send private personal messages (PM) to other forum members
- Customize your profile page and make new friends
Posted 07 March 2011 - 04:38 PM
||...Ads Help To Support This Site...||
Posted 07 March 2011 - 04:46 PM
Posted 07 March 2011 - 05:04 PM
Yes, on the DVR side and if neither then it's on the DECA side, but this is where I doubt a rebooting would always get it back.
Then therefore two possible outcomes - no voltage from DVR and drop (perhaps to 0V) under load, by connecting DECA.
Posted 07 March 2011 - 06:48 PM
All units in DECA cloud
H24-100 removed from service
Posted 07 March 2011 - 07:53 PM
It was an intermittant connection on the short coax into the DECA unit.
The cable is very short and with the DECA unit tucked behind the HR20 and parallel to it, it had just enough stress to make an intermittant connection.
Moved it away from the HR20 so it wasn't parallel and problem solved. Its the cable that comes attached to the DECA unit with the shrink wrap so the assembly was marginal.